Suzhou Xing Yi Machinery Co.Ltd.
Address: Le Yu town of Zhangjiagang City
Chairman: Shi Hezhong
Tel: 0512-58522516 58653980
Hand machine: 013606229802
Zip code: 215622
The letter box:email@example.com
Company address: www.szxyjx.cc
The characteristics of a low-speed refrigerated centrifuge:
The 1 microprocessor control speed, temperature, time and LED digital display or a liquid crystal display (LCD) (user selected).
2 the use of large torque of Brushless DC motor, large starting torque, acceleration / deceleration time is short.
3 with over temperature, over speed, electronic locks, and other safety protection function; with three layers of protective cover, safe and reliable, to ensure personal safety, machine.
The 4 is equipped with a variety of rotor, convenient for the user to choose.
5 Import compressors fluorine free (refrigerant R404a) to meet the requirements of environmental protection.
6 Application of radioimmunoassay and biochemistry, bio pharmaceuticals, blood separation and purification.
Two, the maintenance of low speed refrigerated centrifuge:
On the 1 circuit, electronic circuit and mechanical transmission part of routine inspection and maintenance, pay attention to all kinds of lights is damaged and aging, must be regular maintenance.
2 keep the centrifugal cup symmetrical balance. If the centrifuge chamber placed sample tube is not balanced, will cause a shift jitter centrifuge. Stop after reset, to check whether the normal instrument.
3 centrifuge door bolt, should always check, remain flexible, prevent forced open or closed. When abnormal, to timely replacement of maintenance. The common faults, usually to strengthen the observation, eliminate them timely.
You should check the rotor has no corrosion and fine cracks 4 before use, the rotor has corrosion or crack the prohibition of the use, over the shelf life of the rotor, should always check, guarantee personal safety.
5 after centrifugal centrifugal chamber to dry the water on a regular basis, the motor shaft cone painted a little neutral grease protection. Long time without centrifuge rotors should be removed, cleaned and placed in a dry place.
6 do the centrifuge working state and worked every time the registration work equipment fault record and troubleshooting time, ensure the equipment in the best condition.